DELIVERY AND INSTILATION
The outer packaging is removed and the unit is carefully lifted away from the packaging material. Do not use sharp tools that can scratch the unit.
Please ensure that the following components are present when unpacking the unit.
|1||SANDMAN™ Futura w/ hand piece hoses.|
|1||Foot pedal unit and 4 mm air hose for connection to compressor.|
|1||Hand piece tip ø5, 60º, attached|
|1||Hand piece tip ø6, 80º, attached|
|1||Nozzle cleaner for hand piece|
|1 box||Containing: Futura aluminium oxide powder 3 disposable cups of 29 µm, 3 disposable cups of 45 µm|
|4||Mounting screws type: T-10 torx notch|
|1||DVD with instructions|
|1||16 mm aspirator tube with tip|
Order form for extra angle pieces bits and Futura powder is enclosed.
Installation and assembly
It is important always to take into account the most ergonomic and correct positioning of the SANDMAN™ Futura, so it can be within easy reach of dentist.
If it is not immediately feasible to position the SANDMAN™ Futura unit on the dental chair column by using the three-point swing arm, one should always attempt to sell a floor standing unit with wheels as shown in the illustration.
Position SANDMAN™ Futura in the room where it is to be used.
Ensure prior to installation that the clinic is equipped with an aspirator of normal strength (min. 25 l/min) and that the compressor is able to deliver dry and clean air at a min. 30 l/min to the SANDMAN™ Futura.
Fit the unit in an upright position to the swing arm which should be fitted to a pillar, floor stand or table stand with the matching screws. Then connect the food pedal’s hose to the unit. Air will be supplied via the Press-T quick coupling inside the unit by cutting the air hose behind the throttle// and air filter, so that only dry and clean air is used.
When fitting the unit onto the lamp casing an aesthetic solution is achieved by drilling a hole into the lamp casing behind the bracket and then feeding the air hose through as a hidden installation.
Fit the disposable powder cups in the correct side Left hand side: 29 μm (White lidded cup) – Right hand side: 45 μm (Black lidded cup) or 120 my (white lidded and labelled cup).
Check that the hand pieces supplied and prefitted to the device hoses have been fitted correctly, and ensure that the knob has been set to the hand piece to be tested. Test in a dust cabinet or plastic bag.